The main components of JY-8SK Laser Assisted Steel Bar scanner are:
1) Instrument host
3) Data transfer line
4) Software CD-ROM
5) Bluetooth receiver (optional)
6) Large capacity external power supply (optional)
Hardware features of the instrument:
-High resolution color LCD screen (2.8-inch, 320x240-pixel, 65536-color)
-Integrated host, testing unit integrated with the host, no need for signal line
-Built-in large-capacity lithium battery, the maximum working time can be up to 30 hours
-Small dimension: 22CM × 12CM × 11.8CM
- Laser indicating positioning, convenient to determine the position of steel bar
* The laser beam attached to this instrument meets the safety standards and must not be directly viewed as a laser source
1.2 Performance characteristics
-The testing results show the thickness of the protective layer, the distance between steel bars and the number of steel bars.
-Waveform testing can be carried out to facilitate the analysis of steel bar situation.
-The position of the steel bar and the middle position of the two bars can be accurately positioned by the aiming frame.
-The calibration value is stored automatically, and the calibration is avoided before each test. In special cases, the calibration can be carried out again.
-Thickness testing, section scanning and grid scanning are all boundless, so they can be scanned in a wide range.
-Waveform scan can complete a range of 5.5 meters or more.
-Built-in large capacity memory chip, can store more than 2, 000 components or 170,000 testing points; convenient in the field for data storage, view, deletion.
-Simple data transfer, as long as the instrument boots up (in the main interface, main menu interface, transfer interface) and is connected to the PC with data line, the instrument does not need any operation, the data can be transferred to the computer.
-In the transport interface, data can be transmitted directly through Bluetooth on a computer (optional).
-Supporting professional data analysis software, can test data for professional processing and generate a variety of test reports.
1.4 Usage and maintenance
-Please read this Manual carefully before using the instrument.
-Work environment requirements:
Ambient temperature: -10 ℃ ~ 40 ℃
Relative humidity: < 90%RH
Electromagnetic interference: no strong alternating electromagnetic field
-Storage environment requirements
Make sure the instrument is dry before storage
Ambient temperature: -20 ℃ ~ 50 ℃
Relative humidity: < 90%RH
-Avoid the water of the instrument
-Avoid using in strong magnetic field environment, such as large electromagnets, transformers, and converters, etc.
-Use soft cloth to clean the instrument, if necessary, apply a little alcohol and water to the soft cloth.
-Do not open the case of the instrument without permission.